Swipe om te navigeren naar een ander artikel
The use of biplane fluoroscopy to track bones in the foot is challenging, due to distortion, overlap and image artefact inherent in fluoroscopy systems and high speed photography. The accuracy and precision of these systems have been reported [1–4] and are presented here for our biplane fluoroscopy system.
Biplane Fluoroscopy System: The system consists of two Philips BV Pulsera C-arms set in custom frames around a raised floor with a radiolucent imaging area. X-ray images are captured with high speed (1000fps) cameras. Validation Object: 1.6mm tantalum beads were placed in a machined block (wand) then measured to 7 microns with a Coordinate Measuring Machine to determine their centroid location. The wand was translated and rotated via a 1 micron precision stepper-motor for static validation, as well as manually swept through the field of view at ~0.5m/s for dynamic. Static Accuracy and Precision: accuracy was defined as the RMS error between the translation of the stepper-motor and the measured movement of the beads; precision is defined as the standard deviation of the bead locations. For rotation, accuracy was defined as the RMS error between the applied and measured rotation of the wand. Dynamic Accuracy and Precision: accuracy was defined as the RMS error between the known and measured inter-bead distance; precision was the standard deviation of the inter-bead distance. 3D location processing was accomplished using custom software written in MatLab to derive the 3D location of objects from two, time-synchronized, 2D fluoroscopy images of known spatial relationship. This software also compensates for the image distortion (Figure 1).
Translation: the overall RMS error was 0.066 mm, with a precision of ± 0.016 mm. Rotation: the RMS error was 0.125°. Dynamic motion: the overall RMS error was 0.126 mm, with a precision of ± 0.122 mm.
Brainerd EL, et al: X-ray reconstruction of moving morphology (XROMM): precision, accuracy and applications in comparative biomechanics research. J Exp Zool A Ecol Genet Physiol. 2010, 313: 262-279. PubMed
Miranda D, et al: Accuracy and precision of 3-D skeletal motion capture technology. 56th ORS. 2010, Paper no. 334
- Preliminary marker-based validation of a novel biplane fluoroscopy system
Joseph M Iaquinto
David R Haynor
Bruce J Sangeorzan
William R Ledoux
- BioMed Central